Current Region: Global (EN)
CAUTION: RUNNING A TEFLON-TIPPED PLUNGER DRY MAY CAUSE THE PTFE TO SHRED OR PULL LOOSE FROM THE PLUNGER ROD, RESULTING IN LEAKS
Gas tight syringes with a seal prevent the solvent from evaporating, however the fluid mustn’t be ejected quickly (forcefully) as this damages the seal on the plunger because it rubs against the glass barrel. Depending on the task at hand, we offer syringes with and without a seal.
ILS syringes are precisely manufactured. To avoid imprecision when dosing manually do not dose volumes smaller than 10% of the total capacity. For example: no less than 10µl with a 100µl syringe.
To use a needle with 17 degree bevel for example, our type “A”, where the forces required for penetration are minimal. Please check the tip for damage, otherwise it can damage the septa and contaminate the sample with particles and/or clog the needle itself.
For special cases and for use with Headspace we offer needles type “D” tip, which has a closed tip (dome) and a hole on the side.
Use our HPLC syringes with needle tip “B”. These needles are designed specifically for HPLC valves so the seal will not be damaged.
We recommend needles bevel “B”.
You can use our bevel “C”, which we developed specifically for GC Autosamplers. With needles of a gauge 23 and 23s you can dose using Merlin®Ventile. Thanks to the bevel “C” the needle is protected from bending if the needle hits the bottom of the vial, which also prevents damage to the septa.
Our T-Series (plunger in needle) leaves no fluid in the needle after ejection since the plunger ends at the very tip of the needle. The modular design of the T-series allows quick and easy cleaning, replacement of any damaged parts and can even be put together and repaired on the spot by you without any need for qualified technicians or costly shipping back and forth.
Different levels of viscosity require different needles with specific inner diameter. We offer individual advice in that matter, however, a manual is also available in our catalogue and on our website.
By regular flushing, however there mustn’t be any fluid left in the needle after flushing. You can dry it with vacuum by pulling the air through for example with a higher volume syringe or with a vacuum pump. ILS syringes with a small inner diameter of the needle are shipped with a cleaning wire. In addition, a selection of cleaning wires for different diameters is available. If you work with substances which cause minor clogging, take a look at ILS syringes with removable needles.
Perform cleaning procedures often. If some substance remains in the needle you should use our special coated needles.
The N-Series syringes must be flushed and dried often to prevent the plunger from getting stuck.
However, our A-Series ™ with special seal deals with this problem much better and can also handle sticky liquids easily. The plunger goes through a seal on the top end, the movement of the plunger has been upgraded (no side loading), which ensures a longer lifetime for your syringe.
Remember that all glues are synthetic and have a corresponding solvent. For aggressive substances use a syringe with a removable needle. If possible, we can also repair the syringes for a small fee.
ILS glass syringes are manufactured, meeting exact standards and are quality tested to meet or exceed our minimum performance specifications for accuracy and precision. Our testing methods are in accordance with GLP protocols and utilize equipment that is traceable to NIST standards. Copies of actual test results, are available upon request.